DA 2 FORM 1 JUL 79 TEAR ALONG DOTTED LINE BE EXACT PIN POINT WHERE IT IS from MSC at Jacksonville State University. DA Form Hand Receipt/Annex Number. DA Form Equipment Inspection and Maintenance Worksheet DA Form Equipment. Mail your letter, DA Form (Recommended Changes to. Publications and Blank Forms) or DA Form located in back of this manual.
|Genre:||Health and Food|
|Published (Last):||7 December 2017|
|PDF File Size:||13.57 Mb|
|ePub File Size:||9.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
Supply operating power to the radiacmeter. Never attempt to replace a tube in the radiacmeter.
If the Radiac Set has been modified, then an MWO data plate should appear on the case next to the nomenclature plate. Why should they get your money? This manual cannot list all malfunctions that may occur, not all tests for inspections and corrective actions.
If the surface area missing paint is larger than 1 square inch, touch up painting is required. Each component is shown below. The TM HR consists of. Assure that the test sample is securely attached to the radiac set case with the chain provided. Refer to TM for specific instructions on the control, safe handling, inspection, storage, and disposition ea the test sample.
Overhaul is normally the highest degree of maintenance performed by the Army. Case Headset 1 Inspect for dirt, fungus and grease.
Place the radiacmeter in the case. The numbers indicate the applicable tool or test equipment for the maintenance functions.
Administrative storage covers the storage of equipment which can be readied for mission performance within 24 hours. R E P L A C E – The act of substituting a serviceable like type part, sub-assembly, or module component or assembly for an unserviceable counterpart. Liberated Manuals — free army and government manuals Why do I ofrm it? Covered storage is required.
This figure 228-2 the active time required to perform that maintenance function at the indicated category of maintenance. Tighten the screws securely in order to maintain a watertight battery compartment. If a malfunction is not listed or is not corrected by performing listed corrective actions, notify your Supervisor and evacuate to higher maintenance for repair. This manual has a companion document with a TM number followed by. Online form and submit it for 208-2 faster response. Inspect for tears in the cables.
The Radiac Set consists of: The column gives the area of the damaged item. INSTALL – The act of emplacing, seating, or fixing into position an item, part, module component or assembly in a manner to allow the proper functioning of the equipment or system.
ALIGN – To torm specified variable elements of an item to bring about optimum or desired performance. Shielding – Any material or obstruction which absorbs radiation and thus tends to protect personnel or materials from the effects of a nuclear or atomic explosion. Send us an EIR.
Major Components All Components of the radiac set are contained in a carrying case. If the rubber gasket is damaged, then remove the gasket. Refer to the latest issue of DA Pam to determine whether there. T E S T – To verify serviceability and to detect incipient failure by measuring the mechanical or electrical characteristics of an item and comparing those characteristics with prescribed standards.
R E P A I R – The application of maintenance services inspect, test, service, adjust, align, calibrate, replace or other maintenance actions welding, grinding, riveting, straightening, facing, remachining, or resurfacing to restore serviceability to an item by correcting specific damage, fault, malfunction, or failure in a part, subassembly, module component or assemblyend item, or system.
Should you receive spare tubes by mistake, return to stock immediately. Why are not letting you give those free manuals to your friends? Activates batteries and controls sensitivity of meter.